ESA Science & Technology - Publication Archive
The Mid-Infrared Instrument (MIRI) is a 5 to 28 micron imager and spectrometer that is slated to fly aboard the JWST in 2013. Each of the flight arrays is a 1024x1024 pixel Si:As impurity band conductor detector array, developed by RaytheonVision Systems. JPL, in conjunction with the MIRI science team, has selected the three flight arrays along with their spares. We briefly summarize the development of these devices, then describe the measured performance of the flight arrays along with supplemental data from sister flight-like parts.
The James Webb Space Telescope (JWST) mission is a collaborative project between the National Aeronautics and Space Administration (NASA), the European Space Agency (ESA) and the Canadian Space Agency (CSA) and is considered as the successor to the Hubble Space Telescope (HST). The European contribution consists in providing the Ariane 5 launcher and two out of the four instruments: a combined mid-infrared camera/spectrograph (MIRI) and a near infrared spectrograph (NIRSpec). This article will address the mechanical aspects of NIRSpec by providing an overview of the design drivers and the related solutions for the structure, the thermal design and the mechanisms so as to achieve the required stringent optical performances. The industrial set-up and the project development status will also be presented.
The Grating and Filter Wheel Mechanisms of the JWST NIRSpec instrument allow for reconfiguration of the spectrograph in space in a number of NIR sub-bands and spectral resolutions. Challenging requirements need to be met simultaneously including high launch loads, the large temperature shift to cryo-space, high position repeatability and minimum deformation of the mounted optics. The design concept of the NIRSpec wheel mechanisms is based on the ISOPHOT Filter Wheels but with significant enhancements to support much larger optics. A well-balanced set of design parameters was to be found and a considerable effort was spent to adjust the hardware within narrow tolerances.
We present interim results from the characterization test development for the Detector Subsystem of the Near-Infrared Spectrograph (NIRSpec). NIRSpec will be the primary near-infrared spectrograph on the James Webb Space Telescope (JWST). The Detector Subsystem consists of a Focal Plane Assembly containing two Teledyne HAWAII-2RG arrays, two Teledyne SIDECAR cryogenic application specific integrated circuits, and a warm Focal Plane Electronics box. The Detector Characterization Laboratory at NASA's Goddard Space Flight Center will perform the Detector Subsystem characterization tests. In this paper, we update the initial test results obtained with engineering grade components.
We present the development of a Focal Plane Module (FPM) for the Mid-Infrared Instrument on JWST. MIRI will include three FPMs, two for the spectrometer channels and one for the imager channel. The FPMs are designed to support the detectors at an operating temperature of 6.7 K with high temperature stability and precision alignment while being capable of surviving the launch environment. The flight units will be built and will undergo a rigorous test program in the first half of 2008. This paper includes a description of the full test program and will present the results.
Aims: We report on the production of a large area, shallow, sky survey, from XMM-Newton slews. The great collecting area of the mirrors coupled with the high quantum efficiency of the EPIC detectors have made XMM-Newton the most sensitive X-ray observatory flown to date. We use data taken with the EPIC-pn camera during slewing manoeuvres to perform an X-ray survey of the sky.
Methods: Data from 218 slews have been subdivided into small images and source searched. This has been done in three distinct energy bands; a soft (0.2-2 keV) band, a hard (2-12 keV) band and a total XMM-Newton band (0.2-12 keV). Detected sources, have been quality controlled to remove artifacts and a catalogue has been drawn from the remaining sources.
Results: A "full" catalogue, containing 4710 detections and a "clean" catalogue containing 2692 sources have been produced, from 14% of the sky. In the hard X-ray band (2-12 keV) 257 sources are detected in the clean catalogue to a flux limit of
- AO-7 Announcement
- AO-6 Programme
- Science Operations
- Contacting ISOC
We present how it is achieved to mount a double prism in the filter wheel of MIRIM - the imager of JWST's Mid Infrared Instrument. In order to cope with the extreme conditions of the prisms' surroundings, the low resolution double prism assembly (LRSDPA) design makes high demands on manufacturing accuracy. The design and the manufacturing of the mechanical parts are presented here, while 'Manufacturing and verification of ZnS and Ge prisms for the JWST MIRI imager' are described in a second paper. We also give insights on the astronomical possibilities of a sensitive MIR spectrometer. Low resolution prism spectroscopy in the wavelength range from 5-10 microns will allow to spectroscopically determine redshifts of objects close to/at the re-ionization phase of the universe.
One of the main objectives of the instrument MIRI, the Mid-InfraRed Instrument, of the JWST is the direct detection and characterization of extrasolar giant planets. For that purpose, a coronagraphic device including three Four-Quadrant Phase Masks and a Lyot coronagraph working in mid-infrared, has been developed. We present here the results of the first test campaign of the coronagraphic system in the mid-infrared in the facility developed at the CEA. The performances are compared to the expected ones from the coronagraphic simulations. The accuracy of the centering procedures is also evaluated to validate the choice of the on-board centering algorithm.
The MTS, MIRI Telescope Simulator, is developed by INTA as the Spanish contribution of MIRI (Mid InfraRed Instrument) on board JWST (James Web Space Telescope). The MTS is considered as optical equipment which is part of Optical Ground Support Equipment for the AIV/Calibration phase of the instrument at Rutherford Appleton Laboratory, UK. It is an optical simulator of the JWST Telescope, which will provide a diffractionlimited test beam, including the obscuration and mask pattern, in all the MIRI FOV and in all defocusing range. The MTS will have to stand an environment similar to the flight conditions (35K) but using a smaller set-up, typically at lab scales. The MTS will be used to verify MIRI instrument-level tests, based on checking the implementation/realisation of the interfaces and performances, as well as the instrument properties not subject to interface control such as overall transmission of various modes of operation. This paper includes a functional description and a summary of the development status.
The James Webb Space Telescope (JWST) mission is a collaborative project between the National Aeronautics and Space Administration (NASA), the European Space Agency (ESA) and the Canadian Space Agency (CSA). JWST is considered the successor to the Hubble Space Telescope (HST) and although its design and science objectives are quite different, JWST is expected to yield equivalently astonishing breakthroughs in infrared space science. Due to be launched in 2013 from the French Guiana, the JWST observatory will be placed in an orbit around the anti- Sun Earth-Sun Lagrangian point, L2, by an Ariane 5 launcher provided by ESA. The payload on board the JWST observatory consists of four main scientific instruments: a near-infrared camera (NIRCam), a mid-infrared camera/spectrograph (MIRI), a near-infrared tunable filter (TFI) and a near-infrared spectrograph (NIRSpec). The instrument suite is completed by a Fine Guidance Sensor (FGS). NIRSpec is a multi-object spectrograph capable of measuring the spectra of about 100 objects simultaneously at low (R~100), medium (R~1000) and high (R~2700) resolutions over the wavelength range between 0.6 micron and 5.0 micron. It features also a classical fix-slits spectroscopy mode as well as a 3D-spectrography mode with spectral resolutions up to 2700. The availability of extensive and accurate calibration data of the NIRSpec instrument is a key element to ensure that the nominal performance of the instrument will be achieved and that high-quality processed data will be made available to the users. In this context, an on-ground calibration is planned at instrument level that will supplement the later in-flight calibration campaign.
-- the remainder of the abstract was truncated --
Microshutter arrays are one of the novel technologies developed for the James Webb Space Telescope (JWST). It will allow Near Infrared Spectrometer (NIRSpec) to acquire spectra of hundreds of objects simultaneously therefore increasing its efficiency tremendously. We have developed these programmable arrays that are based on Micro-Electro Mechanical Structures (MEMS) technology. The arrays are 2D addressable masks that can operate in cryogenic environment of JWST. Since the primary JWST science requires acquisition of spectra of extremely faint objects, it is important to provide very high contrast of the open to closed shutters. This high contrast is necessary to eliminate any possible contamination and confusion in the acquired spectra by unwanted objects. We have developed and built a test system for the microshutter array functional and optical characterization. This system is capable of measuring the contrast of the mciroshutter array both in visible and infrared light of the NIRSpec wavelength range while the arrays are in their working cryogenic environment. We have measured contrast ratio of several microshutter arrays and demonstrated that they satisfy and in many cases far exceed the NIRSpec contrast requirement value of 2000.
MIRI is the mid-IR instrument for the James Webb Space Telescope and provides imaging, coronography and integral field spectroscopy over the 5-28 micron wavelength range. MIRI is the only instrument which is cooled to 7K by a dedicated cooler, much lower than the passively cooled 40K of the rest of JWST, which introduces unique challenges. The paper will describe the key features of the overall instrument design. The flight model design of the MIRI Optical System is completed, with hardware now in manufacture across Europe and the USA, while the MIRI Cooler System is at PDR level development. A brief description of how the different development stages of the optical and cooling systems are accommodated is provided, but the paper largely describes progress with the MIRI Optical System. We report the current status of the development and provide an overview of the results from the qualification and test programme.
The Mid-Infrared Instrument (MIRI) is one of the three scientific instruments to fly on the James Webb Space Telescope (JWST), which is due for launch in 2013. MIRI contains two sub-instruments, an imager, which has low resolution spectroscopy and coronagraphic capabilities in addition to imaging, and a medium resolution IFU spectrometer. A verification model of MIRI was assembled in 2007 and a cold test campaign was conducted between November 2007 and February 2008. This model was the first scientifically representative model, allowing a first assessment to be made of the performance. This paper describes the test facility and testing done. It also reports on the first results from this test campaign.